Time clock setting
How It Works
Resetting the time is done by placing a time file on the SD card, you set this file for 1 or 2 minutes into the future
allowing you time to get the SD card back into the SteriLogger and restart the sterilizer at the set time.
The SteriLogger will then read the time file from the SD card and set it’s internal clock accordingly.
This is to be done only if the SteriLogger’s internal clock needs to be set for your local time zone.
Do not include daylight saving time. The time needs to be set to your local standard time.
If you are currently in daylight saving time (summer time) set the time file back 1 hour.
Be sure to include the extra minute or two to complete this task.
Instructions for Setting the
Internal Clock on the Sterilogger
Take the SD card from the SteriLogger and put into your computer. Use the tool below to generate a timestamp.
Set the time a minute or two ahead of your current time. Click convert.
Remember to subtract 1 hour if you’re in daylight saving time!
Open a copy of notepad.
Now just copy and paste the time stamp number into notepad.
In notepad click ‘File’ then ‘Save As’
Make sure the SD card is highlighted, click once on it, if needed.
Name the file timeset.cfg
Below the Name box, Set the file type to all files and save it to the DL-05 SD card.
Eject the SD card from the computer properly. Put the SD card back into the SteriLogger.
Then unplug the sterilizer and plug it back in at the time you set with this tool.
Use 24 hour clock only!
SteriScan™ System Install
Video showing the install of the SteriScan System.